Taking advantage of our spindle motor design experiences, we develop and manufacture many magnet analyzers with high accuracy, functionality, and operability.

We also produce a magnet analyzer which fulfills each customer's needs.

Ideal for magnetic vector analysis of sensor magnet

In FSMA-1500, it is necessary to measure magnetic flux density in X, Y, Z three axis directions in magnetic measurement of a single magnet not a closed magnetic path such as a sensor magnet. We developed a planar type 3-axis magnet analyzer pursuing accuracy, operability, and analyticity. It is ideal for magnetic vector analysis of sensor magnet.

<Plane vector measurement waveform image (3D)>

- Measurement moving range X: 150mm Y: 150mm Z:150 mm
- Moving minimum resolution X:0.01mm Y:0.01mm Z:0.01mm
- Measurement magnetic flux range 0.01mT (millitesla) to 4T (tesla)
- Measurement position of true value by encoder
- High-speed sampling (about 8 times faster than that of normal type)

- The industry's first function 1

The measured vector data waveform can be displayed while rotating 360° using the mouse. - The industry's first function 2

Detailed data of an arbitrary point can be displayed by clicking the vector waveform shape. - The world's smallest size

Ultrafine probe FC-075 can be used (width 0.6mm × thickness 0.28mm) - Simple operation

Fine adjustment of slider position by JOG dial is possible (moving minimum resolution 0.01 mm).

Magnet analyzer/measuring waveform image

- Easy positioning of slider by jog dial

0.01mm/0.05mm/0.1mm step - Memory function and program measurement of X, Y, Z coordinates are possible.
- Point auto function:
- Frequency analysis function (patented)
- Skew angle measurement is possible (patented).
- Rotational angle measurement. max.43,200p/rev
- Up to 200 mm for automatic movement and up to 0.01 mm for resolution are possible.
- 2D editing function (2D)
- Data transfer to spreadsheet software:
- ASCII output calculation
- Vector measurement/analysis (optional)
- The ultrafine probe F-075 (width 0.6 mm × thickness 0.28 mm) can be used.

- Various vector map display:
- Synthesis waveform display in two axis directions (three axis directions) is possible.
- The industry's first

angle display function of synthesis waveform and frequency analysis

This has a 3-D display; X=rotation angle (Deg), Y=Surface magnetic-flux density (Tesla), Z=height (mm)

This also has the frequency 3D display; X=Frequency (Hz), Y=magnetic-flux (dB), Z=direction of height (mm)

This can be used as a 2D display device. It is also possible to pick-up data from a 3D ripple to a 2D ripple.

Automatic pointing function is optional.

- Frequency analysis function (patented)
- This can measure skew angle patented
- Rotation angle resolution: max. 43,200p/rev
- Automatic transfer: max. 200mm, Resolution: max. 0.01mm
- 2D editing function (2 dimensional)
- This can forward the data to a spreadsheet software.
- ASCII output function
- Measurement and analysis of vectors (optional)
- Automatic pointing function (optional)
- The ultra superfine probe F-075 (width 0.6mm × thick 0.28mm) can be used.

- Frequency analysis function patented
- Rotation angle resolution: max. 10,000p/rev
- This can compare the data to a SIN ripple and display the difference.
- The data of maximum value, angle, and area of each pole,
- Measurement at any points is possible with the horizontal cursor.
- This can display the data of the close list with the vertical cursor.
- It is possible to transfer the ripple to any points.
- This can display 6 samples at the same time with the memory function.
- The ultra superfine probe F-075 (width 0.6mm × thick 0.28mm) can be used.

*The specification and the exterior might be changed without notice.